Varicella-Zoster Virus, Molecular Detection, PCR
Method Description Describes how the test is performed and provides a method-specific reference
Viral nucleic acid is extracted by the MagNA Pure automated instrument (Roche Applied Science) from clinical specimens. Primers directed to target DNA (ss DNA binding proteins [gene 29]) produce a 202 bp amplicon. The LightCycler instrument amplifies and monitors by fluorescence the development of target nucleic acid sequences after the annealing step during PCR cycling. This is an automated PCR system that can rapidly detect (30-40 minutes) amplicon development though stringent air-controlled temperature cycling in capillary cuvettes. The detection of amplified products is based on the fluorescence resonance energy transfer (FRET) hybridization probe with a donor fluorophore, fluorescein, on the 3' end is excited by an external light source and emits light that is absorbed by a second hybridization probe with an acceptor fluorophore, LC-Red 640, at the 5' end. The acceptor fluorophore then emits a light of a different wavelength that can be measured with a signal that is proportional to the amount of specific PCR product. Melting curve analysis is performed following PCR amplification. Starting at 45 degrees C, the temperature in the thermal chamber is slowly raised to 80 degrees C, and the fluorescence is measured at frequent intervals. Analysis of the PCR amplification and probe melting curves is accomplished through the use of LightCycler software.(Espy MJ, Teo R, Ross TK, et al: Diagnosis of varicella-zoster virus infections in the clinical laboratory by LightCycler PCR. J Clin Microbiol 2000;38:3187-3189)
Supplemental Report Indicates whether the report includes an additional document with charts, images or other enriched information
Day(s) and Time(s) Test Performed Outlines the days and times the test is performed. This field reflects the day and time the sample must be in the testing laboratory to begin the testing process and includes any specimen preparation and processing time required before the test is performed. Some tests are listed as continuously performed, which means assays are performed several times during the day.
Monday through Saturday; varies
Analytic Time Defines the amount of time it takes the laboratory to setup and perform the test. This is defined in number of days. The shortest interval of time expressed is "same day/1 day," which means the results may be available the same day that the sample is received in the testing laboratory. One day means results are available 1 day after the sample is received in the laboratory.
1 day/same day
Maximum Laboratory Time Defines the maximum time from specimen receipt at Mayo Medical Laboratories until the release of the test result
Specimen Retention Time Outlines the length of time after testing that a specimen is kept in the laboratory before it is discarded
Performing Laboratory Location The location of the laboratory that performs the test