Hepatitis Be Antigen, Serum
Method Description Describes how the test is performed and provides a method-specific reference
This test is performed using the FDA-approved VITROS HBeAg Reagent Pack and the Immunodiagnostic Product HBeAg Calibrator on the VITROS ECi/ECiQ Immunodiagnostic System based on chemiluminescence immunoassay principle. An immunometric technique is used. This involves the simultaneous reaction of HBeAg in the sample with biotinylated mouse monoclonal HBeAg antibody and horseradish peroxidase (HRP)-labeled mouse monoclonal HBeAg antibody in the conjugate. The immune complex is captured by streptavidin on the wells, unbound materials are removed by washing. The bound HRP conjugate is measured by a luminescent reaction. A reagent containing luminogenic substrates (a luminol derivative and a peracid salt) and an electron transfer agent, is added to the wells. The HRP in the bound conjugate catalyzes the oxidation of the luminol derivative, producing light. The electron transfer agent (a substituted acetanilide) increases the level of light produced and prolongs its emission. The light signals are read by the system. The amount of HRP conjugate bound is indicative of the level of HBeAg present in the sample.(Package insert: VITROS Immunodiagnostic Product HBeAg Reagent Pack, No. GEM1222, version 1.0; Ortho-Clinical Diagnostics, Rochester, NY 14626-5101, 5/24/2011)
PDF Report Indicates whether the report includes an additional document with charts, images or other enriched information
Day(s) and Time(s) Test Performed Outlines the days and times the test is performed. This field reflects the day and time the sample must be in the testing laboratory to begin the testing process and includes any specimen preparation and processing time required before the test is performed. Some tests are listed as continuously performed, which means assays are performed several times during the day.
Monday through Saturday
Analytic Time Defines the amount of time it takes the laboratory to setup and perform the test. This is defined in number of days. The shortest interval of time expressed is "same day/1 day," which means the results may be available the same day that the sample is received in the testing laboratory. One day means results are available 1 day after the sample is received in the laboratory.
Maximum Laboratory Time Defines the maximum time from specimen receipt at Mayo Medical Laboratories until the release of the test result
Specimen Retention Time Outlines the length of time after testing that a specimen is kept in the laboratory before it is discarded
Performing Laboratory Location The location of the laboratory that performs the test