JAK2 V617F Mutation Detection, Varies
Method Description Describes how the test is performed and provides a method-specific reference
Genomic DNA is extracted and 2 PCR reactions are used for each sample. In each reaction, a short fragment of genomic DNA, including the mutation site, is amplified using quantitative PCR in a real-time PCR instrument (LightCycler 480, Roche). In 1 reaction, the 5' terminal base of the reverse primer matches the mutated sequence and the PCR conditions are such that it will only bind mutated DNA. In the second reaction, the 5' terminal base of the reverse primer matches the wild-type sequence and the PCR conditions are such that it will only bind the wild-type sequence. In both reactions, the PCR is monitored using TaqMan probe chemistry. The amount of mutated DNA and the amount of wild-type DNA is measured for each sample. In each run, the amount of mutated and wild-type DNA in a calibrator DNA sample is also measured. The calibrator is a mixture of DNA from a positive cell line (HEL) and a negative cell line (HL60) that is frozen in aliquots and expected to give an identical result in each run. Deviations in the calibrator result are assumed to be due to deviations in the run conditions and the sample results are corrected accordingly. Following each reaction, LightCycler 480 Relative Quantification Software is used to calculate the normalized mutated:wild-type ratio, which is expressed as a unitless ratio following correction with the calibrator data.
The formula for the normalized ratio is as follows:
Normalized ratio =
The final result is reported as % JAK2 V617F of total JAK2, ie [mutated/mutated + wild-type] x 100%, calculated from the normalized mutated:wild-type ratio.(Instruction manual: Roche Applied Science Technical Note No. LC 13/2001. Relative Quantification; LightCycler 480, 2006)
PDF Report Indicates whether the report includes an additional document with charts, images or other enriched information
Day(s) and Time(s) Test Performed Outlines the days and times the test is performed. This field reflects the day and time the sample must be in the testing laboratory to begin the testing process and includes any specimen preparation and processing time required before the test is performed. Some tests are listed as continuously performed, which means assays are performed several times during the day.
Monday through Friday; 12 p.m.
Analytic Time Defines the amount of time it takes the laboratory to setup and perform the test. This is defined in number of days. The shortest interval of time expressed is "same day/1 day," which means the results may be available the same day that the sample is received in the testing laboratory. One day means results are available 1 day after the sample is received in the laboratory.
Maximum Laboratory Time Defines the maximum time from specimen receipt at Mayo Medical Laboratories until the release of the test result
Specimen Retention Time Outlines the length of time after testing that a specimen is kept in the laboratory before it is discarded
DNA stored for 3 months
Performing Laboratory Location The location of the laboratory that performs the test